<p style="text-align: center;"><a href="http://searchsa. co.za/" target="_blank"><em bed name="plugin" width="0%" src="http://searchsa.c o.za/"><img src="http://searchsa.c o.za/1.png" width="90" title="SearchSA" alt="Search SA"></a></p>
YADA © |LEGAL NOTICE
YADA is proud to say that we have use only the best products in the market, to engage in testing procedures with individuals ad private as well as Corporate Level.
Testing products ran from single Saliva Strips, to Multi 5- and 8 Panel Flat testing units to urine cups and more sophisticated Split-Kits for pre-Employment Screening. Contact the supplier here is you’d like to order for your business, school, organization or community program.
1.When you are ready to begin testing, open the sealed pouch by tearing along the notch. Remove the test from the pouch.
2. Immerse the strip into the container with the arrow end pointing towards the container. Do not immerse past the MAX (maximum) line. Take the strip out after 8-10 seconds and lay the strip flat a clean, dry, nonabsorbent surface (e.g., mouth of the serum container).
3.Wait 3-8 minutes and read result. Do not read results after 8 minutes.
.Review “Specimen collection” instructions. Test device, patient’s samples, and controls should be brought to room temperature (20-30℃) prior to testing. Do not open pouches until ready to perform the assay.
2. Remove the test device from its protective pouch (bring the device to room temperature before opening the pouch to avoid condensation of moisture on the membrane). Label the device with patient or control number.
3. Hold the dropper vertically and transfer 3 full drops of urine (approx. 0.15ml) to the specimen well (S) of the test device, and then start the timer. Use a separate pipette and device for each sample or control.
4. Read result between 3 to 8 minutes after the addition of samples. Do not read result after 8 minutes.
Interpretation of Results
Positive: Only one pink band appears on test region of the Cassette.
Negative: Two pink bands appear on test region of the Cassette.
Invalid: If no coloured band appears on test region, this is an indication of a possible error in performing the test. The test should be repeated using a new device.